Variations in Cannabinoid Reporting: Part Two
Based on a presentation to the American Chemical Society Fall Conference, 2016
by Savino Sguera
Calibration and Reference Standards
Instrument calibration is the process of generating what is called a calibration curve. A calibration curve is a slope that relates the instruments response to established values.
When an instrument is first installed, its response to chemicals needs calibration. The instrument is calibrated by running a series of standards, chemicals with a precisely known concentration. These series of standards produce a slope that sets the instruments’ response to known amounts of the chemical being analyzed. At this point, an unknown concentration can be run through the instrument and can compared against the slope to pinpoint the precise concentration of chemicals in the unknown sample. This is how all client samples are run by an analytical lab, as all client samples are treated as a sample of unknown concentration until the laboratory releases their report on them.
If calibration is performed properly, a laboratory’s reporting of an unknown sample will be very accurate. But if the calibration is slightly off, the potential to report incorrect values for unknown solutions becomes very high.
The main source of erroneous calibrations are the standards. If a lab has an accurate standard, it will generate an accurate curve. If this standard is higher in concentration than the label value of the standard, it will generate a curve that skews unknowns to lower values. If the standard is lower in concentration than the label value of the standard, it will generate a curve that skews unknowns to higher values.
An example is illustrated in Figure 2. The black line represents a calibration curve created with no errors. The black curve is generated off a standard labeled at 1000 ug/ml and with a true value of 1000 ug/ml. When an “unknown” standard with a true value of 800 ug/ml is tested against the black curve, it generates a value of 800 ug/ml.
Next look at the red calibration curve. This curve was generated using a standard which was labeled at 1000 ug/ml but with a true value of 750 ug/ml. This results in an erroneous calibration curve, and when an “unknown” sample with a true value of 800 ug/ml is run against it, it reports a value of 1067 ug/ml. That’s a 33% increase when compared to the black calibration curve. The red curve’s standard was 25% lower than the label value of 1000 ug, and caused a 33% increase over the value reported by the black calibration curve.
Standards can be lower than their label value for a variety of reasons, but the source of problem can be determined. A retailer of reference standards typically has certificates of precision for their products, and they can trace the sample back to the manufacturer with ease by lot and batch number. Most discrepancies between the true value and label value of a standard occur after the standard has been shipped from the manufacturer to the client laboratory. It is up to the laboratory to ensure that the standard is correct and consistent with previous standards in addition to being consistent with the same standard from a second manufacturer.
The most common reason that a standard would be at a lower concentration than the label value is due to degradation. THCA degrades above room temperature to THC easily. Labs can prevent this degradation by ensuring that standards are delivered to laboratories on on dry ice to prevent heat based degradation, and stored in a -20 C freezer.
Editor’s Note: Standards may eventually degrade even when used under appropriate conditions! Make sure that the standard is replaced and tested against on a regular basis.
Other sources of standard manipulation may be intentional. By adding excess solvent, even a drop, to a certified reference standard, a dishonest person can lower the concentration of chemicals in the standard through dilution. This increases the reported results and is very difficult to detect. Some labs may attempt to manipulate their standards and then attempt to use standard manufacturers’ certificates of precision as a legal defense. Though this manipulation can be very difficult to detect, a skilled administrator can discern when manipulation is present by cross-checking the manufacturer’s certificates against the laboratories result and as well by cross-checking the results of one lab with those of others for the same sample.
Women in the Cannabis Industry: Stay StrongSeptember 20, 2018
Who’s Buying All That Pot? A Look at the Demographics of Cannabis ConsumersSeptember 19, 2018
How to Grow Cannabis 142 – How to Sex Your PlantsSeptember 18, 2018
How to Grow Cannabis 219 – Integrated Pest ManagementSeptember 17, 2018
Do you want to receive the next Grower’s Spotlight as soon as it’s available? Sign up below!
- Want to learn more about subjects touched upon in this article? Check out our articles on subjects such as:
- Inside Look at Leading AZ Extraction Company
- Variations in Cannabinoid Reporting Part One
- Treatise on Decarboxylation: Part One
- Treatise on Decarboxylation: Part Two
- Smoking vs Eating Cannabis: The Effects on Patient Health: Part One
- Smoking vs Eating Cannabis: The Effects on Patient Health: Part Two
- Smoking vs Eating Cannabis: The Effects on Patient Health: Part Three
- Want to get in touch with Savino? He can be reached via the following methods:
Do you have any questions or comments?
About the Author
Savino Sguera is founder and CSO of Digamma Consulting. Since 2010 he has been an analytical chemist and researcher in the cannabis industry, working with both private and public interests to bring scientific integrity to the business. Savino holds a B.Sci. in Biomedical Engineering from Columbia University.